Track topics on Twitter Track topics that are important to you
Abstract Background: The novel urine chemistry analyzer iChem Velocity (IRIS Diagnostics) offers improved urinalysis automation options through integration with the well-established iQ200 urine microscopy analyzer. In the course of optimizing the workflow in our hospital routine laboratory, we evaluated the performance of the iChem Velocity. Methods: A total of 257 random urine samples were analyzed with the iChem Velocity, iQ200, Clinitek Atlas (Siemens Healthcare Diagnostics) and by manual microscopy. Results: Depending on the parameter, 93% (hemoglobin) to 100% (urobilinogen), the iChem Velocity and Clinitek Atlas results agreed within the same rank or within one level of difference. The Clinitek Atlas featured a higher sensitivity for hemoglobin (area under the curve 0.86) and leukocyte esterase (area under the curve 0.85) compared with the iChem Velocity (area under the curve for hemoglobin 0.73, leukocytes 0.78). Imprecision was highest for hemoglobin and leukocytes in a pathological sample pool. While the precision of the Clinitek Atlas for hemoglobin measurements was superior, the iChem Velocity was more precise in analyzing protein and pH. Conclusions: Through urinalysis automation with the iChem Velocity and iQ200, we achieved a reduction of hands-on time by 89%. The sensitivity of this new system should be further improved through ongoing development.
Charité University Medicine Berlin, Institute of Laboratory Medicine and Pathobiochemistry, Berlin, Germany.
This article was published in the following journal.
Name: Clinical chemistry and laboratory medicine : CCLM / FESCC
A chemistry point-of-care analyzer would be useful for evaluating injured wildlife, particularly White rhinoceros (Ceratotherium simum) that survive poaching attempts. The IDEXX VetTest could be suita...
For the medico-psychological assessment (MPA) during driving license re-granting in Germany, abstinence control including urine samples is required. In these programs, even small amounts of markers fo...
Methods for counting residual leukocytes in leukocyte-depleted plasma-a comparison between a routine hematology instrument, the Nageotte chamber, flow cytometry, and a fluorescent microscopy analyzer.
Counting very low levels of leukocytes is technically challenging but mandatory for quality control of leukocyte-depleted plasma. Established assays, such as flow cytometry and counting in the Nageott...
This investigation examined differences in functional sit-to-stand power/velocity between cohorts of adults aged 18-97 years. This study included 264 healthy adults classified into 4 cohorts (18-40, C...
Accurate evaluation of hematology analyzers is recommended before these devices can be broadly introduced for the routine testing of continuous ambulatory peritoneal dialysis (CAPD), ascitic, and pleu...
This is an interim report for POC testing of skyla Clinical Chemistry Analyzer and four skyla Test Systems (ALB, BUN, GLU, and TP). The analyses are performed with skyla Clinical Chemistry...
The SONOCLOT Analyzer (S1; Sienco, Inc. Boulder, CO USA, Model DP-2951) is used to perform kaolin-based activated clotting time as routine monitor of heparin therapy in patients undergoing...
The objectives of the Healthy.io Method Comparison and User Performance Study are: 1. To evaluate the performance of Dip.io Device in comparison to the ACON U500 Mission® U500 Uri...
Previous reports have the evidence implicating gait velocity may affect kinematic parameters. However, how the gait velocity will influence kinematic parameters remain unclear. The purpose...
The human cornea is affected by the magnitude and velocity of both internal and external forces because the cornea has both static and dynamic resistance components. Considering these natu...
The specialty of ANALYTIC CHEMISTRY applied to assays of physiologically important substances found in blood, urine, tissues, and other biological fluids for the purpose of aiding the physician in making a diagnosis or following therapy.
Diagnostic procedures, such as laboratory tests and x-rays, routinely performed on all individuals or specified categories of individuals in a specified situation, e.g., patients being admitted to the hospital. These include routine tests administered to neonates.
A mass spectrometry technique using two (MS/MS) or more mass analyzers. With two in tandem, the precursor ions are mass-selected by a first mass analyzer, and focused into a collision region where they are then fragmented into product ions which are then characterized by a second mass analyzer. A variety of techniques are used to separate the compounds, ionize them, and introduce them to the first mass analyzer. For example, for in GC-MS/MS, GAS CHROMATOGRAPHY-MASS SPECTROMETRY is involved in separating relatively small compounds by GAS CHROMATOGRAPHY prior to injecting them into an ionization chamber for the mass selection.
Measurement of the index of refraction (the ratio of the velocity of light or other radiation in the first of two media to its velocity in the second as it passes from one into the other).
The expenses incurred by a hospital in providing care. The hospital costs attributed to a particular patient care episode include the direct costs plus an appropriate proportion of the overhead for administration, personnel, building maintenance, equipment, etc. Hospital costs are one of the factors which determine HOSPITAL CHARGES (the price the hospital sets for its services).
Benign Prostatic Hyperplasia (BPH) Erectile Dysfunction Urology Urology is the branch of medicine concerned with the urinary tract and diseases that affect it. Examples include urethritis, urethrostenosis and incontinence. Urology is a su...
Health care (or healthcare) is the diagnosis, treatment, and prevention of disease, illness, injury, and other physical and mental impairments in humans. Health care is delivered by practitioners in medicine, chiropractic, dentistry, nursing, pharmacy, a...
preventative maintenance for "i chem" velocityichem velocity vs ichemvelocity urine analyzeriris ichem velocity manualbad ichem velocity strips false positive bloodIris iChem Velocity Stripsichem velocity methodichem velocityiCHEM Velocitywhy cant we run bloody urine on velocityiris ichem velocityiris ichem velocity operation manualichem velocity strip vs multistixIris iChemVELOCITY MANUALvelocity ua testing