Track topics on Twitter Track topics that are important to you
Abstract Background: The novel urine chemistry analyzer iChem Velocity (IRIS Diagnostics) offers improved urinalysis automation options through integration with the well-established iQ200 urine microscopy analyzer. In the course of optimizing the workflow in our hospital routine laboratory, we evaluated the performance of the iChem Velocity. Methods: A total of 257 random urine samples were analyzed with the iChem Velocity, iQ200, Clinitek Atlas (Siemens Healthcare Diagnostics) and by manual microscopy. Results: Depending on the parameter, 93% (hemoglobin) to 100% (urobilinogen), the iChem Velocity and Clinitek Atlas results agreed within the same rank or within one level of difference. The Clinitek Atlas featured a higher sensitivity for hemoglobin (area under the curve 0.86) and leukocyte esterase (area under the curve 0.85) compared with the iChem Velocity (area under the curve for hemoglobin 0.73, leukocytes 0.78). Imprecision was highest for hemoglobin and leukocytes in a pathological sample pool. While the precision of the Clinitek Atlas for hemoglobin measurements was superior, the iChem Velocity was more precise in analyzing protein and pH. Conclusions: Through urinalysis automation with the iChem Velocity and iQ200, we achieved a reduction of hands-on time by 89%. The sensitivity of this new system should be further improved through ongoing development.
Charité University Medicine Berlin, Institute of Laboratory Medicine and Pathobiochemistry, Berlin, Germany.
This article was published in the following journal.
Name: Clinical chemistry and laboratory medicine : CCLM / FESCC
Glucose meters are used routinely in hospital wards to manage blood glucose levels in patients requiring frequent monitoring of blood glucose.
Variation in different urinary measurements for evaluation of iodine status is of concern to clinicians and researchers.
OBJECTIVE To compare blood glucose concentrations of black-tailed prairie dogs (Cynomys ludovicianus) measured by use of a variety of portable analyzers with results from a laboratory biochemistry ana...
The emergency departments (EDs) of Chinese hospitals are gradually being equipped with blood gas machines. These machines, along with the measurement of biochemical markers by the hospital laboratory,...
Rapid evaluation of fibrinogen (Fbg) levels is essential for maintaining homeostasis in patients with massive bleeding during severe trauma and major surgery. This study evaluated the accuracy of fibr...
The SONOCLOT Analyzer (S1; Sienco, Inc. Boulder, CO USA, Model DP-2951) is used to perform kaolin-based activated clotting time as routine monitor of heparin therapy in patients undergoing...
There is an increasing awareness of the importance of treating procedure-related pain. Patients undergoing vascular access procedures are often afraid of needles and the discomfort associa...
We intend to study children and adolescents from 8 to 18 years suffering from diabetes mellitus type 1 for more than one year. The patients will undergo a detailed clinical examination for...
The purpose of this research study is examine the response of the blood vessels in the eye to a light flicker stimulus using the Dynamic Vessel Analyzer (DVA) on two separate days in healt...
Annualised height velocity after 12/24 months treatment and HV SDS height velocity after 12/24 months treatment expressed as number of standard deviations difference from the mean populati...
The specialty of ANALYTIC CHEMISTRY applied to assays of physiologically important substances found in blood, urine, tissues, and other biological fluids for the purpose of aiding the physician in making a diagnosis or following therapy.
Diagnostic procedures, such as laboratory tests and x-rays, routinely performed on all individuals or specified categories of individuals in a specified situation, e.g., patients being admitted to the hospital. These include routine tests administered to neonates.
A mass spectrometry technique using two (MS/MS) or more mass analyzers. With two in tandem, the precursor ions are mass-selected by a first mass analyzer, and focused into a collision region where they are then fragmented into product ions which are then characterized by a second mass analyzer. A variety of techniques are used to separate the compounds, ionize them, and introduce them to the first mass analyzer. For example, for in GC-MS/MS, GAS CHROMATOGRAPHY-MASS SPECTROMETRY is involved in separating relatively small compounds by GAS CHROMATOGRAPHY prior to injecting them into an ionization chamber for the mass selection.
Measurement of the index of refraction (the ratio of the velocity of light or other radiation in the first of two media to its velocity in the second as it passes from one into the other).
The expenses incurred by a hospital in providing care. The hospital costs attributed to a particular patient care episode include the direct costs plus an appropriate proportion of the overhead for administration, personnel, building maintenance, equipment, etc. Hospital costs are one of the factors which determine HOSPITAL CHARGES (the price the hospital sets for its services).
Benign Prostatic Hyperplasia (BPH) Erectile Dysfunction Urology Urology is the branch of medicine concerned with the urinary tract and diseases that affect it. Examples include urethritis, urethrostenosis and incontinence. Urology is a su...
Health care (or healthcare) is the diagnosis, treatment, and prevention of disease, illness, injury, and other physical and mental impairments in humans. Health care is delivered by practitioners in medicine, chiropractic, dentistry, nursing, pharmacy, a...
ichem velocityICHEM VELOCITY STRIPS CHARTichem velocity chemistry strips valuesiChem Velocity Procedureiris velocity performancevelocity chemistry stripsIchem Velocity ProcedureIchem Velocity StripsCOM on Iris Velocityhow to connect urine velocity to networKvelocity urine analyzerichem velocity maintenance logIS ICHEM VELOCITY IS SAME AS ICHEM100Ichem Velocity Urine Chemistry Stripsiris cem velocity