Track topics on Twitter Track topics that are important to you
We demonstrate electron microscopy of fully hydrated eukaryotic cells with nanometer resolution. Living Schizosaccaromyces pombe cells were loaded in a microfluidic chamber and imaged in liquid with scanning transmission electron microscopy (STEM). The native intracellular (ultra)structures of wild-type cells and three different mutants were studied without prior labeling, fixation, or staining. The STEM images revealed various intracellular components that were identified on the basis of their shape, size, location, and mass density. The maximal achieved spatial resolution in this initial study was 32 ± 8 nm, an order of magnitude better than achievable with light microscopy on pristine cells. Light-microscopy images of the same samples were correlated with the corresponding electron-microscopy images. Achieving synergy between the capabilities of light and electron microscopy, we anticipate that liquid STEM will be broadly applied to explore the ultrastructure of live cells.
Department of Molecular Physiology and Biophysics, Vanderbilt University School of Medicine, Nashville, Tennessee.
This article was published in the following journal.
Name: Biophysical journal
The knowledge of the absorption spectrum of hydrated electron is of importance in numerous pulse radiolysis studies because it is often used for dosimetry or for second-order rate constant determinati...
The composition and surface properties of atmospheric aerosol particles largely control their impact on climate by affecting their ability to uptake water, react heterogeneously, and nucleate ice in c...
The preparation of biological cells for either scanning or transmission electron microscopy requires a complex process of fixation, dehydration and drying. Critical point drying is commonly used for s...
The properties of the hydrated electron at the air/water interface are computed for both a cavity and a non-cavity model using mixed quantum/classical molecular dynamics simulation. We take advantage ...
A new form of WO2Cl2 was obtained by modification of a literature procedure. Both the newly prepared WO2Cl2 and the commercial yellow WO2Cl2 exhibited an orthorhombic structure (powder X-ray diffracti...
Comparison of dusting rate of bronchoalveolar lavage in healthy volunteers and patients suffering from sarcoidosis. Blind analysis of bronchoalveolar lavage with Optical Microscopy, Trans...
RATIONALE: Evaluating patients who have skin diseases may help doctors plan more effective treatment and follow-up for skin diseases and systemic diseases that show up in the skin. PURPOS...
RATIONALE: Viral therapy may be able to kill tumor cells without damaging normal cells. PURPOSE: This phase II trial is studying the side effects and how well viral therapy works in treat...
A large body of evidence confirm the cholesterol lowering effect of phytosterols and red yeast rice. Because their mechanisms of action mime the ones of chemical statins and cholesterol ab...
To evaluate the pharmacokinetic (PK) and pharmacodynamic (PD) response of a hydrated and non-hydrated 84 mg octreotide implant in patients with acromegaly in the first 6 weeks of treatment...
Microscopy using an electron beam, instead of light, to visualize the sample, thereby allowing much greater magnification. The interactions of ELECTRONS with specimens are used to provide information about the fine structure of that specimen. In TRANSMISSION ELECTRON MICROSCOPY the reactions of the electrons that are transmitted through the specimen are imaged. In SCANNING ELECTRON MICROSCOPY an electron beam falls at a non-normal angle on the specimen and the image is derived from the reactions occurring above the plane of the specimen.
Electron microscopy in which the ELECTRONS or their reaction products that pass down through the specimen are imaged below the plane of the specimen.
Electron microscopy involving rapid freezing of the samples. The imaging of frozen-hydrated molecules and organelles permits the best possible resolution closest to the living state, free of chemical fixatives or stains.
A type of TRANSMISSION ELECTRON MICROSCOPY in which the object is examined directly by an extremely narrow electron beam scanning the specimen point-by-point and using the reactions of the electrons that are transmitted through the specimen to create the image. It should not be confused with SCANNING ELECTRON MICROSCOPY.
Microscopy in which the object is examined directly by an electron beam scanning the specimen point-by-point. The image is constructed by detecting the products of specimen interactions that are projected above the plane of the sample, such as backscattered electrons. Although SCANNING TRANSMISSION ELECTRON MICROSCOPY also scans the specimen point by point with the electron beam, the image is constructed by detecting the electrons, or their interaction products that are transmitted through the sample plane, so that is a form of TRANSMISSION ELECTRON MICROSCOPY.