How Important is the Recommended Slow Cuff Pressure Deflation Rate for Blood Pressure Measurement?
Summary of "How Important is the Recommended Slow Cuff Pressure Deflation Rate for Blood Pressure Measurement?"
Cuff pressure deflation rate influences blood pressure (BP) measurement. However, there is little quantitative clinical evidence on its effect. Oscillometric pulses recorded from 75 subjects at the recommended deflation rate of 2-3 mmHg per second were analyzed. Some pulses were removed to realize six faster rates (2-7 times faster than the original). Systolic, diastolic, and mean arterial blood pressures (SBP, DBP, MAP) were determined from the original and six reconstructed oscillometric waveforms. Manual measurement was based on the appearance of oscillometric pulse peaks, and automatic measurement on two model envelopes (linear and polynomial) fitted to the sequence of oscillometric pulse amplitudes. The effects of deflation rate on BP determination and within-subject BP variability were analyzed. For SBP and DBP determined from the manual measurement, different deflation rates resulted in significant changes (both p < 0.001). However, for SBP, DBP, and MAP determined from the automatic linear and polynomial model techniques, there was no deflation rate effect (all p > 0.3). Faster deflation increased the within-subject BP variability (all p < 0.001). In conclusion, for the manual technique accurate BP measurement could be achieved only with the recommended slow deflation rate, and for the automatic model-based techniques, the deflation rate had little effect.
Cardiovascular Physics and Engineering Research Group, Regional Medical Physics Department, Newcastle University, Freeman Hospital, Newcastle upon Tyne, NE7 7DN, UK, firstname.lastname@example.org.
This article was published in the following journal.
Name: Annals of biomedical engineering
- PubMed Source: http://www.ncbi.nlm.nih.gov/pubmed/21735319
- DOI: http://dx.doi.org/10.1007/s10439-011-0347-9
Medical and Biotech [MESH] Definitions
Instruments for measuring arterial blood pressure consisting of an inflatable cuff, inflating bulb, and a gauge showing the blood pressure. (Stedman, 26th ed)
The signs of life that may be monitored or measured, namely pulse rate, respiratory rate, body temperature, and blood pressure.
An alpha-2 adrenergic agonist that crosses the blood-brain barrier. Clonidine acts centrally by reducing sympathetic tone, resulting in a fall in diastolic and systolic blood pressure and a reduction in heart rate. It also acts peripherally, and this peripheral activity may be responsible for the transient increase in blood pressure seen during rapid intravenous administration. (From Martindale, the Extra Pharmacopoeia, 30th ed, p350)
Blood Pressure Monitoring, Ambulatory
Method in which repeated blood pressure readings are made while the patient undergoes normal daily activities. It allows quantitative analysis of the high blood pressure load over time, can help distinguish between types of HYPERTENSION, and can assess the effectiveness of antihypertensive therapy.
Persistently high systemic arterial BLOOD PRESSURE. Based on multiple readings (BLOOD PRESSURE DETERMINATION), hypertension is currently defined as when SYSTOLIC PRESSURE is consistently greater than 140 mm Hg or when DIASTOLIC PRESSURE is consistently 90 mm Hg or more.
Oscillometric determination of blood pressure may be advantageous, as cuff inflation requires lower cuff pressure and shorter duration than deflation. In this observational study, we compared the bloo...
We observed an increase in mean middle cerebral artery blood flow velocity (V (mca)) after tourniquet deflation during orthopedic surgery under sevoflurane anesthesia in patients with diabetes mellitu...
Blood pressure cuff dimensions directly affect accuracy of blood pressure measurements. For accurate blood pressure measurements, the width of the cuff must be proportional to arm circumference. A cuf...
A host of animal studies have been used to model the effects of exposure to a low protein diet in utero on adult blood pressure. Collection of systolic blood pressure data by the indirect tail-cuff pl...
Cuff-occluded rate of rise of peripheral venous pressure has been proposed to reflect volume changes in experimental studies. The aim of this study was to evaluate changes in cuff-occluded rate of ris...
The purpose of this study is to monitor the blood pressure level of the patient using a cuff ranged 22 cm-30 cm.
The purpose of this study is to monitor the blood pressure level of the patient using a wrist cuff ranged 14 cm-25 cm.
The purpose of this study is to monitor the blood pressure level of the patient using a preformed cuff ranged 22 cm-36 cm.
The purpose of this study is to determine whether deflating the tracheal cuff in tracheostomized patients, during disconnections from ventilatory support along the weaning phase shortens t...
Angioplasty is a commonly performed procedure for treating blockage of the blood vessels in the heart. Although this procedure is usually successful, it is often accompanied by a small deg...