Enhancement of the detection limit for lateral flow immunoassays: Evaluation and comparison of bioconjugates.
Summary of "Enhancement of the detection limit for lateral flow immunoassays: Evaluation and comparison of bioconjugates."
There is an increasing demand for convenient and accurate point-of-care tools that can detect and diagnose different stages of a disease in remote or impoverished settings. In recent years, lateral flow immunoassays (LFIA) have been indicated as a suitable medical diagnostic tool for these environments because they require little or no sample preparation, provide rapid and reliable results with no electronic components and thus can be manufactured at low costs and operated by unskilled personnel. However, even though they have been successfully applied to acute and chronic disease detection, LFIA based on gold nanoparticles, the standard marker, show serious limitations when high sensitivity is needed, such as early stage disease detection. Moreover, based on the lack of comparative information for label performance, significant optimization of the systems that are currently in use might be possible. To this end, in the presented work, we compare the detection limit between the four most used labels: colloidal-gold, silver enhanced gold, blue latex bead and carbon black nanoparticles. Preliminary results were obtained by using the biotin-streptavidin coupling as a model system and showed that carbon black had a remarkably low detection limit of 0.01μg/mL in comparison to 0.1μg/mL, 1μg/mL and 1mg/mL for silver-coated gold nanoparticles, gold nanoparticles and polystyrene beads, respectively. Therefore, as a proof of concept, carbon black was used in a detection system for Dengue fever. This was achieved by immobilizing monoclonal antibodies for the nonstructural glycoprotein (NS1) of the Dengue virus to carbon black. We found that the colorimetric detection limit of 57ng/mL for carbon black was ten times lower than the 575ng/mL observed for standard gold nanoparticles; which makes it sensitive enough to diagnose a patient on the first days of infection. We therefore conclude that, careful screening of detection labels should be performed as a necessary step during LFIA development in order to enhance the detection limit in a final test system.
Helmholtz Zentrum München, German Research Center for Environmental Health, Ingolstädter Landstrasse 1, 85764 Neuherberg, Germany.
This article was published in the following journal.
Name: Journal of immunological methods
- PubMed Source: http://www.ncbi.nlm.nih.gov/pubmed/22100215
- DOI: http://dx.doi.org/10.1016/j.jim.2011.11.003
We developed and validated a novel lateral flow assay for the detection of crustacean protein in processed foods. This assay had high sensitivity; the visual detection limit for shrimp protein extract...
For devices such as bio-/chemical sensors in microfluidic systems, flow fluctuations result in noise in the sensor output. Here, we demonstrate in-line monitoring of flow fluctuations with a cantileve...
Previous studies on circulating microparticles (MPs) indicate that the majority of MPs are of a size below the detection limit of most standard flow cytometers. The objective of the present study was...
Malaria is largely a preventable and curable disease. However, a delay or an inappropriate treatment can result in serious adverse outcome for patient. Rapid, simple and cost-effective diagnostic test...
Existing technologies for analysis of microbiological contaminants in food or clinical samples are often expensive and require laboratory settings and trained personnel. Here we present a lateral flow...
Caudal epidural block is useful not only for regional anesthesia of low abdominal, perineal area or lower extremities but for relieving acute or chronic pain of low back and lower extremit...
This observational study is supposed to assess (under conditions of clinical practice in daily routine) whether treatment with Viramune (nevirapine) in combination with Truvada (tenofovir...
This observational study is supposed to assess (under conditions of clinical practice in daily routine) whether treatment with Viramune (nevirapine) in combination with ARV, e.g. Combivir...
Prospective study in HIV-1 infected patients with a plasma viral load below the limit of detection and stable for at least 5 years.
The purpose of this study is to collect clinical specimens and corresponding clinical data to develop a non-invasive test for detection of intra-amniotic infection and prediction of preter...
Medical and Biotech [MESH] Definitions
Comparison of various psychological, sociological, or cultural factors in order to assess the similarities or diversities occurring in two or more different cultures or societies.
Concentration or quantity that is derived from the smallest measure that can be detected with reasonable certainty for a given analytical procedure.
The use of fluorescence spectrometry to obtain quantitative results for the FLUORESCENT ANTIBODY TECHNIQUE. One advantage over the other methods (e.g., radioimmunoassay) is its extreme sensitivity, with a detection limit on the order of tenths of microgram/liter.
Detection of drugs that have been abused, overused, or misused, including legal and illegal drugs. Urine screening is the usual method of detection.
Studies determining the effectiveness or value of processes, personnel, and equipment, or the material on conducting such studies. For drugs and devices, CLINICAL TRIALS AS TOPIC; DRUG EVALUATION; and DRUG EVALUATION, PRECLINICAL are available.