Track topics on Twitter Track topics that are important to you
We describe a portable gas sensor based on cataluminescence. Miniaturization of the gas sensor was achieved by using a miniature photomultiplier tube, a miniature gas pump and a simple light seal. The signal to noise ratio (SNR) was considered as the evaluation criteria for the design and testing of the sensor. The main source of noise was from thermal background. Optimal working temperature and flow rate were determined experimentally from the viewpoint of improvement in SNR. A series of parameters related to analytical performance was estimated. The limitation of detection of the sensor was 7 ppm (SNR = 3) for ethanol and 10 ppm (SNR = 3) for hydrogen sulphide. Zirconia and barium carbonate were respectively selected as nano-sized catalysts for ethanol and hydrogen sulphide. Copyright © 2012 John Wiley & Sons, Ltd.
State Key Laboratory of Precision Measurement Technology and Instruments, Department of Precision Instruments and Mechanology, Tsinghua University, Beijing, 100084, PR China.
This article was published in the following journal.
Name: Luminescence : the journal of biological and chemical luminescence
A portable laser photoacoustic sensor for methane (CH₄) detection based on a field-programmable gate array (FPGA) is reported. A tunable distributed feedback (DFB) diode laser in the 1654 nm wavelen...
Hypoxia-based cell culture experiments are routine and essential components of in vitro cancer research. Most laboratories use low-cost portable modular chambers to achieve hypoxic conditions for cell...
A portable, recyclable and highly selective paper-based sensor device for the colorimetric and optical detection of hydrogen sulfate anions in water was developed. The detection system features a rhod...
A compact and portable optofluidic microresonator has been fabricated and characterized. It is based on a Fabry-Perot microcavity consisting essentially of two tailored dichroic Bragg mirrors prepared...
Sensitive detection of trimethylamine both in aqueous and gaseous phases has been accomplished using an inexpensive colorimetric sensor array. Distinctive color change patterns provide facile discrimi...
The purpose of this study is to assess the performance of the subcutaneous glucose sensor over an extended sensor life. The sensor is currently approved for 3 days of use and this study wi...
The purpose of this study is to assess performance of a new subcutaneous glucose sensor over a seven day sensor life when used with currently marketed Medtronic Diabetes devices. In additi...
The purpose of the study is to evaluate the safety and effectiveness of the Abbott Sensor Based Glucose Monitoring System Personal (System-P) and Abbott Sensor Based Glucose Monitoring Sys...
Sleep is a fundamental need for human and is associated with the working performance and the disease occurrence. Furthermore, the amount of people with sleep disorder or apnea increased la...
Objectives: The objective of this study is to determine if any differences exist between the varying modes of portable oxygen delivery systems including liquid oxygen, a portable concentr...
Portable or implantable devices for infusion of insulin. Includes open-loop systems which may be patient-operated or controlled by a pre-set program and are designed for constant delivery of small quantities of insulin, increased during food ingestion, and closed-loop systems which deliver quantities of insulin automatically based on an electronic glucose sensor.
A family of intracellular calcium-sensing proteins found predominately in NEURONS and PHOTORECEPTOR CELLS. They contain EF HAND MOTIFS and undergo conformational changes upon calcium-binding. Neuronal calcium-sensor proteins interact with other regulatory proteins to mediate physiological responses to a change in intracellular calcium concentration.
Portable electronics device for storing and playing audio and or media files. MP3 for MPEG-1 audio layer 3, is a digital coding format.
A neuronal calcium sensor protein that is expressed as several isoforms and can interact with ACTIN; TUBULIN; and CLATHRIN.
MICROCOMPUTERS, sometimes called PDA, that are very small and portable, fitting in a hand, and that have much more function than a calculator. They are convenient to use in clinical and other field situations for quick data management. They usually require docking with MICROCOMPUTERS for updates.