Track topics on Twitter Track topics that are important to you
We describe a portable gas sensor based on cataluminescence. Miniaturization of the gas sensor was achieved by using a miniature photomultiplier tube, a miniature gas pump and a simple light seal. The signal to noise ratio (SNR) was considered as the evaluation criteria for the design and testing of the sensor. The main source of noise was from thermal background. Optimal working temperature and flow rate were determined experimentally from the viewpoint of improvement in SNR. A series of parameters related to analytical performance was estimated. The limitation of detection of the sensor was 7 ppm (SNR = 3) for ethanol and 10 ppm (SNR = 3) for hydrogen sulphide. Zirconia and barium carbonate were respectively selected as nano-sized catalysts for ethanol and hydrogen sulphide. Copyright © 2012 John Wiley & Sons, Ltd.
State Key Laboratory of Precision Measurement Technology and Instruments, Department of Precision Instruments and Mechanology, Tsinghua University, Beijing, 100084, PR China.
This article was published in the following journal.
Name: Luminescence : the journal of biological and chemical luminescence
A gaseous sensor system was developed for the detection of methane based on its cataluminescence emission. Cataluminescence characteristics and optimal conditions were studied in detail under optimize...
In this paper, we demonstrate for the first time the use of silver nanoparticles (AgNPs) for colorimetric ascorbic acid (AA) quantification in a paper-based sensor. This device is constituted by spot ...
Melamine detection in milk and other foods has attracted much attention since the discovery that melamine-adulterated food causes severe kidney damage. Although many methods have been developed to det...
Real-time monitoring of specific gas concentrations with a compact and portable gas sensing device is required to sense potential health risk and danger from toxic gases. For such purposes, we develop...
In this article, we consider the design of a human gesture recognition system based on pattern recognition of signatures from a portable smart radar sensor. Powered by AAA batteries, the smart radar s...
The purpose of this study is to assess the performance of the subcutaneous glucose sensor over an extended sensor life. The sensor is currently approved for 3 days of use and this study wi...
The purpose of this study is to assess performance of a new subcutaneous glucose sensor over a seven day sensor life when used with currently marketed Medtronic Diabetes devices. In additi...
The purpose of the study is to evaluate the safety and effectiveness of the Abbott Sensor Based Glucose Monitoring System Personal (System-P) and Abbott Sensor Based Glucose Monitoring Sys...
The APPROPRIATE study will compare differences in functional capacity (peak VO2) between chronotropically incompetent patients randomized to receive rate responsive pacing driven by either...
Objectives: The objective of this study is to determine if any differences exist between the varying modes of portable oxygen delivery systems including liquid oxygen, a portable concentr...
Portable or implantable devices for infusion of insulin. Includes open-loop systems which may be patient-operated or controlled by a pre-set program and are designed for constant delivery of small quantities of insulin, increased during food ingestion, and closed-loop systems which deliver quantities of insulin automatically based on an electronic glucose sensor.
A family of intracellular calcium-sensing proteins found predominately in NEURONS and PHOTORECEPTOR CELLS. They contain EF HAND MOTIFS and undergo conformational changes upon calcium-binding. Neuronal calcium-sensor proteins interact with other regulatory proteins to mediate physiological responses to a change in intracellular calcium concentration.
Portable electronics device for storing and playing audio and or media files. MP3 for MPEG-1 audio layer 3, is a digital coding format.
A neuronal calcium sensor protein that is expressed as several isoforms and can interact with ACTIN; TUBULIN; and CLATHRIN.
MICROCOMPUTERS, sometimes called PDA, that are very small and portable, fitting in a hand, and that have much more function than a calculator. They are convenient to use in clinical and other field situations for quick data management. They usually require docking with MICROCOMPUTERS for updates.