Near-Infrared Laser Desorption/Ionization Aerosol Mass Spectrometry for Investigating Primary and Secondary Organic Aerosols under Low Loading Conditions (dagger).
Summary of "Near-Infrared Laser Desorption/Ionization Aerosol Mass Spectrometry for Investigating Primary and Secondary Organic Aerosols under Low Loading Conditions (dagger)."
A new method, near-infrared laser desorption/ionization aerosol mass spectrometry (NIR-LDI-AMS), is described for the real time analysis of organic aerosols at atmospherically relevant mass loadings. Use of a single NIR laser pulse to vaporize and ionize particle components deposited on an aluminum probe results in minimal fragmentation to produce exclusively intact pseudomolecular anions at [M - H](-). Limits of detection (total particulate mass sampled) for oxidized compounds of relevance to atmospheric primary and secondary organic aerosol range from 89 fg for pinic acid to 8.8 pg for cholesterol. NIR-LDI-AMS was used in conjunction with the University of Vermont Environmental Chamber to study secondary organic aerosol (SOA) formation from ozonolysis of limonene at total aerosol mass loadings ranging from 3.2 to 25.0 mug m(-3) and with a time resolution of several minutes. NIR-LDI-AMS permitted direct delineation between gas-phase, homogeneous SOA formation and subsequent heterogeneous aerosol processing by ozone.
Department of Chemistry, University of Vermont, Burlington, Vermont 05452.
This article was published in the following journal.
Name: Analytical chemistry
Medical and Biotech [MESH] Definitions
Spectrometry, Mass, Matrix-assisted Laser Desorption-ionization
A mass spectrometric technique that is used for the analysis of large biomolecules. Analyte molecules are embedded in an excess matrix of small organic molecules that show a high resonant absorption at the laser wavelength used. The matrix absorbs the laser energy, thus inducing a soft disintegration of the sample-matrix mixture into free (gas phase) matrix and analyte molecules and molecular ions. In general, only molecular ions of the analyte molecules are produced, and almost no fragmentation occurs. This makes the method well suited for molecular weight determinations and mixture analysis.
Tandem Mass Spectrometry
A mass spectrometry technique using two (MS/MS) or more mass analyzers. With two in tandem, the precursor ions are mass-selected by a first mass analyzer, and focused into a collision region where they are then fragmented into product ions which are then characterized by a second mass analyzer. A variety of techniques are used to separate the compounds, ionize them, and introduce them to the first mass analyzer. For example, for in GC-MS/MS, GAS CHROMATOGRAPHY-MASS SPECTROMETRY is involved in separating relatively small compounds by GAS CHROMATOGRAPHY prior to injecting them into an ionization chamber for the mass selection.
Spectrometry, Mass, Electrospray Ionization
A mass spectrometry technique used for analysis of nonvolatile compounds such as proteins and macromolecules. The technique involves preparing electrically charged droplets from analyte molecules dissolved in solvent. The electrically charged droplets enter a vacuum chamber where the solvent is evaporated. Evaporation of solvent reduces the droplet size, thereby increasing the coulombic repulsion within the droplet. As the charged droplets get smaller, the excess charge within them causes them to disintegrate and release analyte molecules. The volatilized analyte molecules are then analyzed by mass spectrometry.
Gas Chromatography-mass Spectrometry
A microanalytical technique combining mass spectrometry and gas chromatography for the qualitative as well as quantitative determinations of compounds.
An analytical method used in determining the identity of a chemical based on its mass using mass analyzers/mass spectrometers.
Matrix-assisted laser desorption/ionization mass spectrometry is an established soft ionization method that is widely applied to analyze biomolecules. The UV-absorbing organic matrix is essential for...
In this work, the combined use of desorption by a continuous wave near infrared diode laser and ionization by a dielectric barrier discharge-based probe (laser desorption dielectric barrier discharge...
Continuous flow infrared matrix-assisted laser desorption electrospray ionization (CF IR MALDESI) mass spectrometry was demonstrated for the on-line analysis of liquid samples. Samples in aqueous solu...
We evaluated whether direct tissue matrix-assisted laser desorption/ionization (MALDI) mass spectrometry (MS) analysis of lipids may distinguish intrahepatic cholangiocarcinomas from adjacent normal t...
The ionization mechanisms of several atmospheric pressure ion sources based on desorption and ionization of samples deposited on a surface were studied. Home-built desorption electrospray ionization (...
RATIONALE: Drugs used in chemotherapy, such as doxorubicin, epirubicin, and cyclophosphamide, work in different ways to stop the growth of tumor cells, either by killing the cells or by st...
RATIONALE: Studying samples of blood from patients with cancer in the laboratory may help doctors learn more about changes that occur and identify biomarkers related to cancer. PURPOSE: T...
We are collecting brain tissue specimens and blood samples from patients at Vanderbilt University Medical Center who are undergoing intracranial surgery to remove brain tissue, including b...
This phase II trial is studying how well decitabine works in treating patients with previously untreated acute myeloid leukemia. Drugs used in chemotherapy, such as decitabine, work in dif...
RATIONALE: Finding specific proteins in the blood may help doctors tell whether a patient has ovarian cancer. PURPOSE: This clinical trial is studying how well proteomic profiling works i...