Track topics on Twitter Track topics that are important to you
High resolution imaging capabilities are essential for accurately guiding successful endovascular interventional procedures. Present x-ray imaging detectors are not always adequate due to their inherent limitations. The newly-developed high-resolution micro-angiographic fluoroscope (MAF-CCD) detector has demonstrated excellent clinical image quality; however, further improvement in performance and physical design may be possible using CMOS sensors. We have thus calculated the theoretical performance of two proposed CMOS detectors which may be used as a successor to the MAF. The proposed detectors have a 300 μm thick HL-type CsI phosphor, a 50 μm-pixel CMOS sensor with and without a variable gain light image intensifier (LII), and are designated MAF-CMOS-LII and MAF-CMOS, respectively. For the performance evaluation, linear cascade modeling was used. The detector imaging chains were divided into individual stages characterized by one of the basic processes (quantum gain, binomial selection, stochastic and deterministic blurring, additive noise). Ranges of readout noise and exposure were used to calculate the detectors' MTF and DQE. The MAF-CMOS showed slightly better MTF than the MAF-CMOS-LII, but the MAF-CMOS-LII showed far better DQE, especially for lower exposures. The proposed detectors can have improved MTF and DQE compared with the present high resolution MAF detector. The performance of the MAF-CMOS is excellent for the angiography exposure range; however it is limited at fluoroscopic levels due to additive instrumentation noise. The MAF-CMOS-LII, having the advantage of the variable LII gain, can overcome the noise limitation and hence may perform exceptionally for the full range of required exposures; however, it is more complex and hence more expensive.
This article was published in the following journal.
Name: Proceedings - Society of Photo-Optical Instrumentation Engineers
High-resolution, low-noise x-ray detectors based on the complementary metal-oxide-semiconductor (CMOS) active pixel sensor (APS) technology have been developed and proposed for digital breast tomosynt...
2D semiconductor materials are being considered for next generation electronic device application such as thin-film transistors and complementary metal-oxide-semiconductor (CMOS) circuit due to their ...
The purpose of this study is to develop a method to evaluate the clinical utility of a high resolution 75µ pixel MAF-CMOS image receptor compared to a 200µ flat-panel detector (FPD).
In optoacoustic tomography, negatively focused transducers may be used for improving the tangential image resolution while preserving a high signal-to-noise ratio. Commonly, image reconstruction in su...
For dedicated brain PET, we can significantly improve sensitivity for the cerebrum region by arranging detectors in a compact hemisphere. The geometrical sensitivity for the top region of the hemisphe...
The purpose of this clinical study is to evaluate the imaging performance of the Carestream PRO 3543 Detector and Carestream PRO 3543C Detector (referred to as the "investigational devices...
This study is designed to evaluate the contribution of multimodal high-resolution EEG-NIRS electrical and local haemodynamic analysis (HR-EEG, HR-NIRS) to spatial localization of epileptog...
Multicenter prospective study comparing the diagnosis value of high-resolution EEG and depth-EEG to localize the epileptogenic zone in drug resistant partial epilepsies.
This study will inform the feasibility of the High Resolution Impedance Manometry (HRIM) system combined with Automated Impedance Manometry (AIM) analysis AIM analysis as a screening tool ...
Objective Sedation is a depression of consciousness varies between anxiolysis and general anesthesia. It is used to facilitate diagnostic tests and minor surgeries. Respiratory complicatio...
A method of chemical analysis based on the detection of characteristic radionuclides following a nuclear bombardment. It is also known as radioactivity analysis. (McGraw-Hill Dictionary of Scientific and Technical Terms, 4th ed)
The identification, analysis, and resolution of moral problems that arise in the care of patients. (Bioethics Thesaurus)
A performance test based on forced MOTOR ACTIVITY on a rotating rod, usually by a rodent. Parameters include the riding time (seconds) or endurance. Test is used to evaluate balance and coordination of the subjects, particular in experimental animal models for neurological disorders and drug effects.
Analysis based on the mathematical function first formulated by Jean-Baptiste-Joseph Fourier in 1807. The function, known as the Fourier transform, describes the sinusoidal pattern of any fluctuating pattern in the physical world in terms of its amplitude and its phase. It has broad applications in biomedicine, e.g., analysis of the x-ray crystallography data pivotal in identifying the double helical nature of DNA and in analysis of other molecules, including viruses, and the modified back-projection algorithm universally used in computerized tomography imaging, etc. (From Segen, The Dictionary of Modern Medicine, 1992)
Theoretical representations that simulate the behavior or activity of systems, processes, or phenomena. They include the use of mathematical equations, computers, and other electronic equipment.
Radiology is the branch of medicine that studies imaging of the body; X-ray (basic, angiography, barium swallows), ultrasound, MRI, CT and PET. These imaging techniques can be used to diagnose, but also to treat a range of conditions, by allowing visuali...