Track topics on Twitter Track topics that are important to you
Failing single-ventricle (SV) patients might benefit from ventricular assist devices (VADs) as a bridge to heart transplantation. Considering the complex physiopathology of SV patients and the lack of established experience, the aim of this work was to realize and test a lumped parameter model of the cardiovascular system, able to simulate SV hemodynamics and VAD implantation effects. Data of 30 SV patients (10 Norwood, 10 Glenn, and 10 Fontan) were retrospectively collected and used to simulate patients' baseline. Then, the effects of VAD implantation were simulated. Additionally, both the effects of ventricular assistance and cavopulmonary assistance were simulated in different pathologic conditions on Fontan patients, including systolic dysfunction, diastolic dysfunction, and pulmonary vascular resistance increment. The model can reproduce patients' baseline well. Simulation results suggest that the implantation of
(i) increases the cardiac output (CO) in all the three palliation conditions (Norwood 77.2%, Glenn 38.6%, and Fontan 17.2%); (ii) decreases the SV external work (SVEW) (Norwood 55%, Glenn 35.6%, and Fontan 41%); (iii) increases the mean pulmonary arterial pressure (Pap) (Norwood 39.7%, Glenn 12.1%, and Fontan 3%). In Fontan circulation, with systolic dysfunction, the left VAD (LVAD) increases CO (35%), while the right VAD (RVAD) determines a decrement of inferior vena cava pressure (Pvci) (39%) with 34% increment of CO. With diastolic dysfunction, the LVAD increases CO (42%) and the RVAD decreases the Pvci. With pulmonary vascular resistance increment, the RVAD allows the highest CO (50%) increment with the highest decrement of Pvci (53%). The single ventricular external work (SVEW) increases (decreases) increasing the VAD speed in cavopulmonary (ventricular) assistance. Numeric models could be helpful in this challenging and innovative field to support patients and VAD selection to optimize the clinical outcome and personalize the therapy.
This article was published in the following journal.
Name: Artificial organs
Right heart failure occurs in 9-44% of left ventricular assist device (LVAD) implants of which less than 10% require a right ventricular assist device (RVAD) support either concurrently with the LVAD ...
To assess baseline patient characteristics and identify factors associated with in-hospital mortality after ventricular assist device (VAD) placement.
Left ventricular assist devices are funded in the UK exclusively as a bridge to transplant (BTT). However, patients who potentially could receive a transplant may develop reversible contraindications ...
Early right ventricular (RV) failure after LVAD increases morbidity and mortality. Percutaneous RVAD (pRVAD) support is an alternative to more invasive surgical RVAD (sRVAD).
Mechanical circulatory support for infants with single ventricle physiology remains challenging. Utilization of a ventricular assist device (VAD) has potential advantages over extracorporeal circulato...
Determination of risk factors of ventricular arrhythmias (VAs) after implantation of continuous flow left ventricular assist device in order to characterize which patient requires ICD impl...
The aim of this study is to determine the safety and efficiency of Remodulin®（Treprostinil Injection）to reduce the pulmonary arterial pressure and prevent pulmonary hypertension (PH) ...
The goal of this project is to develop a patient-centered decision aid for decision-making about advanced heart failure treatment. This study seeks to create a decision aid that presents o...
The purpose of this study is to determine if the delivery of cells just after implantation of left ventricular assist device will help to improve the pumping function of your heart and min...
To retrospectively investigate the effect of physical therapy intervention on improving physical function of patients post left ventricular assist device (LVAD) through electronic medical ...
A potentially lethal cardiac arrhythmia characterized by an extremely rapid, hemodynamically unstable ventricular tachycardia (150-300 beats/min) with a large oscillating sine-wave appearance. If untreated, ventricular flutter typically progresses to VENTRICULAR FIBRILLATION.
Removal of a MEDICAL DEVICE from the market due to the identification of an intrinsic property of the device that results in a serious risk to public health.
Isoforms of MYOSIN TYPE II, specifically found in the ventricular muscle of the HEART. Defects in the genes encoding ventricular myosins result in FAMILIAL HYPERTROPHIC CARDIOMYOPATHY.
An abnormally rapid ventricular rhythm usually in excess of 150 beats per minute. It is generated within the ventricle below the BUNDLE OF HIS, either as autonomic impulse formation or reentrant impulse conduction. Depending on the etiology, onset of ventricular tachycardia can be paroxysmal (sudden) or nonparoxysmal, its wide QRS complexes can be uniform or polymorphic, and the ventricular beating may be independent of the atrial beating (AV dissociation).
A congenital cardiomyopathy that is characterized by infiltration of adipose and fibrous tissue into the RIGHT VENTRICLE wall and loss of myocardial cells. Primary injuries usually are at the free wall of right ventricular and right atria resulting in ventricular and supraventricular arrhythmias.
Cardiology is a specialty of internal medicine. Cardiac electrophysiology : Study of the electrical properties and conduction diseases of the heart. Echocardiography : The use of ultrasound to study the mechanical function/physics of the h...
Organ transplantation is the moving of an organ from one body to another or from a donor site to another location on the patient's own body, for the purpose of replacing the recipient's damaged or absent organ. The emerging field of regenerative ...
Cardiovascular disease (CVD)
Acute Coronary Syndromes (ACS) Blood Cardiovascular Dialysis Hypertension Stent Stroke Vascular Cardiovascular disease (CVD) includes all the diseases of the heart and circulation including coronary heart disease (angina...