Track topics on Twitter Track topics that are important to you
In a surgical operation requiring endoscopy, it is essential to obtain a clear endoscopic view. However, it is often disturbed by the contamination on the lens during the surgery. No device can clean the lens surface simply and completely. Many surgeons are hampered by the impaired view and the distraction by the repeated cleaning of the lens. Therefore, we developed a novel endoscope cleaning device to address this problem. The device was made of 3D-printed rubber-like plastic. It contains a syringe filled with saline and an aspiration system. It would be used intraoperatively to wash the lens surface in a few seconds with rapid flow of water and air. The cleaning ability of the device was evaluated using mayonnaise with adenosine triphosphate (ATP) as a model contaminant. The gauze-wiping maneuver was selected as control. After each maneuver, the clarity of the endoscopic view was evaluated, and residual contaminants were assessed quantitatively with ATP assay. The cleaning device obtained a crisp and clear view and eliminated the contaminant on the lens every time after a single cleaning maneuver. The gauze-wiping maneuver required for the lens to be wiped at least three times to obtain a clear view, and even then, some contaminants remained. Repeated contamination and cleaning using gauze led to accumulation of contaminants on the lens, which resulted in difficulty in cleaning the lens as the operation proceeded. The cleaning device did not show such accumulation. Our novel cleaning device with air and water flow has been shown to wash out the lens contaminants completely and immediately in a simple manner. It is expected to improve the safety and cost-effectiveness of endoscopic surgery.
This article was published in the following journal.
Name: PloS one
Reusable medical devices (RMDs) must be reprocessed between uses to render them safe for each use and each patient. Cleaning used devices removes organic and inorganic soil making them either safe for...
Adequate cleaning of reusable medical devices is critical for preventing cross-infection among patients. For reusable medical devices, cleaning using mechanical brushes and detergent may not be suffic...
Surfaces that stay clean when immersed in water are important for an enormous range of applications from ships and buildings to marine, medical and other equipment. Up till now the main strategy for d...
Disinfection byproduct precursors (DBPs) were removed from raw surface water obtained from two Canadian drinking water treatment plants via adsorption to two regenerable linear engineered TiO nanomate...
A miniature, 10 mm aperture histotripsy transducer with an f-number of 0.7 was fabricated using an elliptically shaped aluminum lens which was epoxy-bonded to an air-backed 5.0 MHz, PTZ-5A, 1-3 dice-a...
Prospective, Randomized Comparison of the Use of FloShield Air System ® Versus the Referenced Technic (Water + Povidone-Iodine Solution) During Gynecologic Endoscopic Surgery to Evaluate the Operative Lens Vision Quality
Endoscopic surgical procedures are becoming more common in all surgical specialties including gynecology that was one of the pioneers in this field specialties. Maintaining a quality fiel...
The objective of this study is to evaluate the safety and effectiveness of NNR06 Multi-Purpose Solution cleaning and disinfecting solution (Test) compared to COMPLETE Multi-Purpose Solutio...
After phacoemulsifying cataract surgery, the cortical fibers of lens remain in the posterior capsule, which can cause early postoperative posterior capsule turbidity and affect the visual ...
To compare patient comfort and overall lens cleanliness with consecutive use of two multipurpose lens solutions.
This study evaluates the addition of an interdental cleaning device paired with a manual toothbrush compared to using a manual toothbrush only. Plaque removal, reduction of gingival bleedi...
The flow of water in enviromental bodies of water such as rivers, oceans, water supplies, aquariums, etc. It includes currents, tides, and waves.
An animal's cleaning and caring for the body surface. This includes preening, the cleaning and oiling of feathers with the bill or of hair with the tongue.
Insertion of an artificial lens to replace the natural CRYSTALLINE LENS after CATARACT EXTRACTION or to supplement the natural lens which is left in place.
Sterile solutions, essentially free from foreign particles and suitably compounded and dispensed, for instillation into the eye. It does not include solutions for cleaning eyeglasses or CONTACT LENS SOLUTIONS. (From Dorland, 27th ed)
The analysis of a chemical substance by inserting a sample into a carrier stream of reagent using a sample injection valve that propels the sample downstream where mixing occurs in a coiled tube, then passes into a flow-through detector and a recorder or other data handling device.
Surgery is a technology consisting of a physical intervention on tissues. All forms of surgery are considered invasive procedures; so-called "noninvasive surgery" usually refers to an excision that does not penetrate the structure being exci...
Food is any substance consumed to provide nutritional support for the body. It is usually of plant or animal origin, and contains essential nutrients, such as carbohydrates, fats, proteins, vitamins, or minerals. The substance is ingested by an organism ...
An assay is an analytic procedure for qualitatively assessing or quantitatively measuring the presence or amount or the functional activity of a target entity. This can be a drug or biochemical substance or a cell in an organism or organic sample. ...