Track topics on Twitter Track topics that are important to you
The need for early identification of postural abnormalities without exposing patients to constant radiation has stimulated the development of instruments aiming to measure the spinal curvatures.
To verify the validity, repeatability and reproducibility of angular measures of sagittal curvatures of the spine obtained using an adapted arcometer, by comparing them with Cobb angles of the respective curvatures obtained by using X-rays.
52 participants were submitted to two procedures designed to evaluate the thoracic and lumbar curvatures: (1) X-ray examination from which the Cobb angles (CA) of both curvatures were obtained, and (2) measuring the angles with the arcometer (AA). Two evaluators collected the data using the arcometer, with the rods placed at T1, T12, L1 and L5 spinous processes levels in a way as to permit linear measurements which, with aid of trigonometry, supplied the AA.
There was a very strong and significant correlation between AA and CA (r=0.94; p<0.01), with no-significant difference (p=0.32), for the thoracic curvature. There was a strong and significant correlation for the lumbar curvature (r=0.71; p<0.01) between AA and CA, with no-significant difference (p=0.30). There is a very strong correlation between intra-evaluator and inter-evaluator AA.
It was possible to quantify reliably the thoracic and lumbar curvatures with the arcometer and it can thus be considered valid and reliable and for use in evaluating spinal curvatures in the sagittal plane.
This article was published in the following journal.
Name: Revista brasileira de fisioterapia (Sao Carlos (Sao Paulo, Brazil))
Lumbar degenerative scoliosis (LDS) is a common spinal disease for senior citizens. However, LDS accompanied with thoracic lordosis and lumbar kyphosis (LK) is rare in clinic. No reports have reported...
To assess the repeatability and reproducibility of Visante™ OCT and Oculus Pentacam HR® in measuring central (CCT) and topographic corneal thickness (TCT) along the principle meridians in keratocon...
In order for a new analytical technique such as surface-enhanced Raman scattering (SERS) to be used in a routine manner, data and studies on the validation of the method are required. In that context,...
In this study, we evaluated the validity of measuring blood pressure (BP) noninvasively in marmosets by using the tail-cuff method. The number of measurements needed for a valid reading was calculated...
Measurement of hemodynamic wall shear stress (WSS) is important in investigating the role of WSS in the initiation and progression of atherosclerosis. Echo particle image velocimetry (echo PIV) is a n...
Evaluate the repeatability and reproducibility of the RTVue-XR for measuring the total corneal thickness (pachymetry), the epithelial thickness, and the stromal thickness mapping in normal...
Evaluate the repeatability and reproducibility of iVue for measuring the total corneal thickness, the epithelial thickness, and the stromal thickness in normal subjects and patients with v...
The purpose of this study is to assess the repeatability and reproducibility of measurements of the vascular structures and the anatomical structures of the posterior pole of the eye based...
The objective of this study is to define the repeatability and reproducibility of quantifying flow-mediated dilation (FMD) of the brachial artery (BA) using the AngioDefender (AD) methodol...
The purpose of this preliminary substudy to the parent study "Aerobic Exercise Intervention for Knee Osteoarthritis" is to determine the repeatability and reproducibility of various measur...
The statistical reproducibility of measurements (often in a clinical context), including the testing of instrumentation or techniques to obtain reproducible results. The concept includes reproducibility of physiological measurements, which may be used to develop rules to assess probability or prognosis, or response to a stimulus; reproducibility of occurrence of a condition; and reproducibility of experimental results.
Determination, by measurement or comparison with a standard, of the correct value of each scale reading on a meter or other measuring instrument; or determination of the settings of a control device that correspond to particular values of voltage, current, frequency or other output.
VERTEBRAE in the region of the lower BACK below the THORACIC VERTEBRAE and above the SACRAL VERTEBRAE.
Tapping of the subarachnoid space in the lumbar region, usually between the third and fourth lumbar vertebrae.
A noninvasive method for assessing BODY COMPOSITION. It is based on the differential absorption of X-RAYS (or GAMMA RAYS) by different tissues such as bone, fat and other soft tissues. The source of (X-ray or gamma-ray) photon beam is generated either from radioisotopes such as GADOLINIUM 153, IODINE 125, or Americanium 241 which emit GAMMA RAYS in the appropriate range; or from an X-ray tube which produces X-RAYS in the desired range. It is primarily used for quantitating BONE MINERAL CONTENT, especially for the diagnosis of OSTEOPOROSIS, and also in measuring BONE MINERALIZATION.
Radiology is the branch of medicine that studies imaging of the body; X-ray (basic, angiography, barium swallows), ultrasound, MRI, CT and PET. These imaging techniques can be used to diagnose, but also to treat a range of conditions, by allowing visuali...