Hardware-efficient low-power image processing system for wireless capsule endoscopy.
Summary of "Hardware-efficient low-power image processing system for wireless capsule endoscopy."
This paper presents the design of a hardware-efficient, low-power image processing system for next-generation wireless endoscopy. The presented system is composed of a custom CMOS image sensor, a dedicated image compressor, a forward error correction (FEC) encoder protecting radio transmitted data against random and burst errors, a radio data transmitter, and a controller supervising all operations of the system. The most significant part of the system is the image compressor. It is based on an integer version of a discrete cosine transform and a novel, low complexity yet efficient, entropy encoder making use of an adaptive Golomb-Rice algorithm instead of Huffman tables. The novel hardware-efficient architecture designed for the presented system enables on-the-fly compression of the acquired image. Instant compression, together with elimination of the necessity of retransmitting erroneously received data by their prior FEC encoding, significantly reduces the size of the required memory in comparison to previous systems. The presented system was prototyped in a single, low-power, 65-nm field programmable gate arrays (FPGA) chip. Its power consumption is low and comparable to other application-specific-integrated-circuits-based systems, despite FPGA-based implementation.
This article was published in the following journal.
Name: IEEE journal of biomedical and health informatics
- PubMed Source: http://www.ncbi.nlm.nih.gov/pubmed/24240723
- DOI: http://dx.doi.org/10.1109/JBHI.2013.2266101
Combining ASIC and multiple microsensors low-power wireless electronic capsule was developed for the long-term monitoring of the entire human gastro-intestinal (GI) tract. To meet the system requireme...
Wireless capsule endoscopy (WCE) is a promising technology for direct diagnosis of the entire small bowel to detect lethal diseases, including cancer and obscure gastrointestinal bleeding (OGIB). To i...
In order to reduce the miss rate of the wireless capsule endoscopy, in this paper, we propose a new system of the endoscopic capsule with multiple cameras. A master-slave architecture, including an ef...
This paper presents the design of a wireless capsule system for monitoring the physiological signals of the human gastrointestinal (GI) tract. The primary components of the system include a wireless c...
In this paper, a new low complexity and lossless image compression system for capsule endoscopy (CE) is presented. The compressor consists of a low-cost YEF color space converter and variable-length p...
The purpose of this study is to conduct a cost-effectiveness analysis of wireless capsule endoscopy in the investigation of patients with overt obscure gastrointestinal bleeding. To inform...
To compare the small bowel cleanliness for wireless capsule endoscopy using two different Polyethylene Glycol administration schedules (before the wireless capsule endoscopy ingestion ver...
Small bowel wireless capsule endoscopy is the investigation modality of choice for suspected diseases of the small bowel. The procedure is safe and noninvasive, the main risk being capsule...
Wireless Capsule Endoscopy has been the most reliable diagnostic standard for small bowel Crohn's disease. Endoscopic healing is an important endpoint to measure response to treatment.
This study compares wireless capsule endoscopy (patients swallow a pill-size camera that sends pictures of the intestine to a recorder worn on their belt) to an x-ray study (called dedicat...
Medical and Biotech [MESH] Definitions
Improvement of the quality of a picture by various techniques, including computer processing, digital filtering, echocardiographic techniques, light and ultrastructural MICROSCOPY, fluorescence spectrometry and microscopy, scintigraphy, and in vitro image processing at the molecular level.
A computer architecture, implementable in either hardware or software, modeled after biological neural networks. Like the biological system in which the processing capability is a result of the interconnection strengths between arrays of nonlinear processing nodes, computerized neural networks, often called perceptrons or multilayer connectionist models, consist of neuron-like units. A homogeneous group of units makes up a layer. These networks are good at pattern recognition. They are adaptive, performing tasks by example, and thus are better for decision-making than are linear learning machines or cluster analysis. They do not require explicit programming.
Improvement in the quality of an x-ray image by use of an intensifying screen, tube, or filter and by optimum exposure techniques. Digital processing methods are often employed.
Communications networks connecting various hardware devices together within or between buildings by means of a continuous cable or voice data telephone system.
A method of delineating blood vessels by subtracting a tissue background image from an image of tissue plus intravascular contrast material that attenuates the X-ray photons. The background image is determined from a digitized image taken a few moments before injection of the contrast material. The resulting angiogram is a high-contrast image of the vessel. This subtraction technique allows extraction of a high-intensity signal from the superimposed background information. The image is thus the result of the differential absorption of X-rays by different tissues.