Hardware-efficient low-power image processing system for wireless capsule endoscopy.
Summary of "Hardware-efficient low-power image processing system for wireless capsule endoscopy."
This paper presents the design of a hardware-efficient, low-power image processing system for next-generation wireless endoscopy. The presented system is composed of a custom CMOS image sensor, a dedicated image compressor, a forward error correction (FEC) encoder protecting radio transmitted data against random and burst errors, a radio data transmitter, and a controller supervising all operations of the system. The most significant part of the system is the image compressor. It is based on an integer version of a discrete cosine transform and a novel, low complexity yet efficient, entropy encoder making use of an adaptive Golomb-Rice algorithm instead of Huffman tables. The novel hardware-efficient architecture designed for the presented system enables on-the-fly compression of the acquired image. Instant compression, together with elimination of the necessity of retransmitting erroneously received data by their prior FEC encoding, significantly reduces the size of the required memory in comparison to previous systems. The presented system was prototyped in a single, low-power, 65-nm field programmable gate arrays (FPGA) chip. Its power consumption is low and comparable to other application-specific-integrated-circuits-based systems, despite FPGA-based implementation.
This article was published in the following journal.
Name: IEEE journal of biomedical and health informatics
- PubMed Source: http://www.ncbi.nlm.nih.gov/pubmed/24240723
- DOI: http://dx.doi.org/10.1109/JBHI.2013.2266101
A wireless motility capsule is a new method for ambulatory assessment of transit times and motility throughout the gastrointestinal tract. The objective of this study was to evaluate the ability of a...
Quality of service (QoS) and, in particular, reliability and a bounded low latency are essential attributes of safety-critical wireless systems for medical applications. However, wireless links are ty...
This article introduces a power-efficient, miniature electronic nose (e-nose) system. The e-nose system primarily comprises two self-developed chips, a multiple-walled carbon nanotube (MWNT)-polymer b...
A body sensor network (BSN) is a wireless network of biosensors and a local processing unit, which is commonly referred to as the personal wireless hub (PWH). Personal health information (PHI) is coll...
The ability to wirelessly power electrical devices is becoming of greater urgency as a component of energy conservation and sustainability efforts. Due to health and safety concerns, most wireless pow...
The purpose of this study is to conduct a cost-effectiveness analysis of wireless capsule endoscopy in the investigation of patients with overt obscure gastrointestinal bleeding. To inform...
Wireless Capsule Endoscopy has been the most reliable diagnostic standard for small bowel Crohn's disease. Endoscopic healing is an important endpoint to measure response to treatment.
This study compares wireless capsule endoscopy (patients swallow a pill-size camera that sends pictures of the intestine to a recorder worn on their belt) to an x-ray study (called dedicat...
To determine if there is a difference between wireless ECG and wired ECG findings.
The purpose of this study is to determine the best role for capsule endoscopy in diagnosing Crohn's disease.
Medical and Biotech [MESH] Definitions
Improvement of the quality of a picture by various techniques, including computer processing, digital filtering, echocardiographic techniques, light and ultrastructural MICROSCOPY, fluorescence spectrometry and microscopy, scintigraphy, and in vitro image processing at the molecular level.
A computer architecture, implementable in either hardware or software, modeled after biological neural networks. Like the biological system in which the processing capability is a result of the interconnection strengths between arrays of nonlinear processing nodes, computerized neural networks, often called perceptrons or multilayer connectionist models, consist of neuron-like units. A homogeneous group of units makes up a layer. These networks are good at pattern recognition. They are adaptive, performing tasks by example, and thus are better for decision-making than are linear learning machines or cluster analysis. They do not require explicit programming.
Improvement in the quality of an x-ray image by use of an intensifying screen, tube, or filter and by optimum exposure techniques. Digital processing methods are often employed.
Communications networks connecting various hardware devices together within or between buildings by means of a continuous cable or voice data telephone system.
A method of delineating blood vessels by subtracting a tissue background image from an image of tissue plus intravascular contrast material that attenuates the X-ray photons. The background image is determined from a digitized image taken a few moments before injection of the contrast material. The resulting angiogram is a high-contrast image of the vessel. This subtraction technique allows extraction of a high-intensity signal from the superimposed background information. The image is thus the result of the differential absorption of X-rays by different tissues.