Hardware-efficient low-power image processing system for wireless capsule endoscopy.
Summary of "Hardware-efficient low-power image processing system for wireless capsule endoscopy."
This paper presents the design of a hardware-efficient, low-power image processing system for next-generation wireless endoscopy. The presented system is composed of a custom CMOS image sensor, a dedicated image compressor, a forward error correction (FEC) encoder protecting radio transmitted data against random and burst errors, a radio data transmitter, and a controller supervising all operations of the system. The most significant part of the system is the image compressor. It is based on an integer version of a discrete cosine transform and a novel, low complexity yet efficient, entropy encoder making use of an adaptive Golomb-Rice algorithm instead of Huffman tables. The novel hardware-efficient architecture designed for the presented system enables on-the-fly compression of the acquired image. Instant compression, together with elimination of the necessity of retransmitting erroneously received data by their prior FEC encoding, significantly reduces the size of the required memory in comparison to previous systems. The presented system was prototyped in a single, low-power, 65-nm field programmable gate arrays (FPGA) chip. Its power consumption is low and comparable to other application-specific-integrated-circuits-based systems, despite FPGA-based implementation.
This article was published in the following journal.
Name: IEEE journal of biomedical and health informatics
- PubMed Source: http://www.ncbi.nlm.nih.gov/pubmed/24240723
- DOI: http://dx.doi.org/10.1109/JBHI.2013.2266101
Wireless capsule endoscopy (WCE) is a promising technology for direct diagnosis of the entire small bowel to detect lethal diseases, including cancer and obscure gastrointestinal bleeding (OGIB). To i...
A power-efficient wireless stimulating system for a head-mounted deep brain stimulator (DBS) is presented. A new adaptive rectifier generates a variable DC supply voltage from a constant AC power carr...
A wireless motility capsule is a new method for ambulatory assessment of transit times and motility throughout the gastrointestinal tract. The objective of this study was to evaluate the ability of a ...
This study investigates the specific absorption rate (SAR) and the in situ electric field in anatomically based human models for the magnetic field from an inductive wireless power transfer system dev...
We introduce a two-loop power control that allows an efficient use of the overall power resources for commercial wireless networks based on cross-layer optimization. This approach maximizes the networ...
The purpose of this study is to conduct a cost-effectiveness analysis of wireless capsule endoscopy in the investigation of patients with overt obscure gastrointestinal bleeding. To inform...
Wireless Capsule Endoscopy has been the most reliable diagnostic standard for small bowel Crohn's disease. Endoscopic healing is an important endpoint to measure response to treatment.
This study compares wireless capsule endoscopy (patients swallow a pill-size camera that sends pictures of the intestine to a recorder worn on their belt) to an x-ray study (called dedicat...
To determine if there is a difference between wireless ECG and wired ECG findings.
The purpose of this study is to determine the best role for capsule endoscopy in diagnosing Crohn's disease.
Medical and Biotech [MESH] Definitions
Improvement of the quality of a picture by various techniques, including computer processing, digital filtering, echocardiographic techniques, light and ultrastructural MICROSCOPY, fluorescence spectrometry and microscopy, scintigraphy, and in vitro image processing at the molecular level.
A computer architecture, implementable in either hardware or software, modeled after biological neural networks. Like the biological system in which the processing capability is a result of the interconnection strengths between arrays of nonlinear processing nodes, computerized neural networks, often called perceptrons or multilayer connectionist models, consist of neuron-like units. A homogeneous group of units makes up a layer. These networks are good at pattern recognition. They are adaptive, performing tasks by example, and thus are better for decision-making than are linear learning machines or cluster analysis. They do not require explicit programming.
Improvement in the quality of an x-ray image by use of an intensifying screen, tube, or filter and by optimum exposure techniques. Digital processing methods are often employed.
Communications networks connecting various hardware devices together within or between buildings by means of a continuous cable or voice data telephone system.
A method of delineating blood vessels by subtracting a tissue background image from an image of tissue plus intravascular contrast material that attenuates the X-ray photons. The background image is determined from a digitized image taken a few moments before injection of the contrast material. The resulting angiogram is a high-contrast image of the vessel. This subtraction technique allows extraction of a high-intensity signal from the superimposed background information. The image is thus the result of the differential absorption of X-rays by different tissues.