Track topics on Twitter Track topics that are important to you
This paper presents the design of a hardware-efficient, low-power image processing system for next-generation wireless endoscopy. The presented system is composed of a custom CMOS image sensor, a dedicated image compressor, a forward error correction (FEC) encoder protecting radio transmitted data against random and burst errors, a radio data transmitter, and a controller supervising all operations of the system. The most significant part of the system is the image compressor. It is based on an integer version of a discrete cosine transform and a novel, low complexity yet efficient, entropy encoder making use of an adaptive Golomb-Rice algorithm instead of Huffman tables. The novel hardware-efficient architecture designed for the presented system enables on-the-fly compression of the acquired image. Instant compression, together with elimination of the necessity of retransmitting erroneously received data by their prior FEC encoding, significantly reduces the size of the required memory in comparison to previous systems. The presented system was prototyped in a single, low-power, 65-nm field programmable gate arrays (FPGA) chip. Its power consumption is low and comparable to other application-specific-integrated-circuits-based systems, despite FPGA-based implementation.
This article was published in the following journal.
Name: IEEE journal of biomedical and health informatics
Infrared fluorescence endoscopy (IRFE), in conjunction with an infrared fluorescent-labelling contrast agent, is a well known technique used for efficient early-stage cancer detection. In this paper w...
The study aimed to develop a real-time electromyography (EMG) signal acquiring and processing device that can acquire signal during electrical stimulation. Since electrical stimulation output can affe...
In wireless positioning systems, the transmitter's power is usually fixed. In this paper, we explore the use of varying transmission powers to increase the performance of a wireless localization syste...
Wireless capsule endoscopy for gastrointestinal (GI) tract is a modern technology that has the potential to replace conventional endoscopy techniques. Capsule endoscopy is a pill-shaped device embedde...
Inspired by ongoing evolutions in the field of wireless body area networks (WBANs), this tutorial paper presents a conceptual and exploratory study of wireless electroencephalography (EEG) sensor netw...
The purpose of this study is to conduct a cost-effectiveness analysis of wireless capsule endoscopy in the investigation of patients with overt obscure gastrointestinal bleeding. To inform...
To compare the small bowel cleanliness for wireless capsule endoscopy using two different Polyethylene Glycol administration schedules (before the wireless capsule endoscopy ingestion ver...
Small bowel wireless capsule endoscopy is the investigation modality of choice for suspected diseases of the small bowel. The procedure is safe and noninvasive, the main risk being capsule...
Wireless Capsule Endoscopy has been the most reliable diagnostic standard for small bowel Crohn's disease. Endoscopic healing is an important endpoint to measure response to treatment.
This study compares wireless capsule endoscopy (patients swallow a pill-size camera that sends pictures of the intestine to a recorder worn on their belt) to an x-ray study (called dedicat...
Improvement of the quality of a picture by various techniques, including computer processing, digital filtering, echocardiographic techniques, light and ultrastructural MICROSCOPY, fluorescence spectrometry and microscopy, scintigraphy, and in vitro image processing at the molecular level.
A computer architecture, implementable in either hardware or software, modeled after biological neural networks. Like the biological system in which the processing capability is a result of the interconnection strengths between arrays of nonlinear processing nodes, computerized neural networks, often called perceptrons or multilayer connectionist models, consist of neuron-like units. A homogeneous group of units makes up a layer. These networks are good at pattern recognition. They are adaptive, performing tasks by example, and thus are better for decision-making than are linear learning machines or cluster analysis. They do not require explicit programming.
Improvement in the quality of an x-ray image by use of an intensifying screen, tube, or filter and by optimum exposure techniques. Digital processing methods are often employed.
Communications networks connecting various hardware devices together within or between buildings by means of a continuous cable or voice data telephone system.
A method of delineating blood vessels by subtracting a tissue background image from an image of tissue plus intravascular contrast material that attenuates the X-ray photons. The background image is determined from a digitized image taken a few moments before injection of the contrast material. The resulting angiogram is a high-contrast image of the vessel. This subtraction technique allows extraction of a high-intensity signal from the superimposed background information. The image is thus the result of the differential absorption of X-rays by different tissues.